Your cart is currently empty!
Jaundice Detector HJD Transcutaneous
- Measurement: light reflective
- LED; Light source: LED
- Indication error:0~25mg/dL± 1.0mg/dL
- Power: Lithium battery(DC3.6V,1000mAh±20%),after a fully charged can check Measuring more than 800 times;
- Check color screen: white end face is 0, the yellow end face 16.0± 1.0
- Can displayed separately mg /dL, umol/L in the meantime.
- Data storage: can storage 200 measuring date.
- Calculating the average function: It has the function that measuring 2-5 times, then workout the
- average automatically.
- Open Preparation time: boot, no need to prepare
- Battery voltage detection function: it will be displayed “Low Battery” when the battery voltage less than 3 .6V.
- Automatic shutdown after the non-working state, for 10 minutes: automatic shutdown function
- Battery display:Top right corner of the viewing screen display
- It respectively means the left battery electric quantity is about 100%,75%,50%,25%,0%.
- charging display: The viewing screen will be lighten and displayed “Charging..” during charging.
- charging automatic protection function: It will stop charging automatically at the time battery voltage charging arrive 4.2V±0.05V.